ALLTEST SARS-CoV-2 rapid antigen detection reagent (nasal swab) provide a simple and fast method for self-testing, especially for asymptomatic infection and early infection detection with high sensitivity and accuracy, Omicron and Delta variant viruses are suitable. Moreover, it is easy to operate, and you can check it out in 15 minutes, allowing you to understand your physical health, protect yourself and your family, and more effectively protect your business and community.
• Simple. Fast. Easy to use and easy to operate
• 15 minutes to test and know
• Highly sensitive and accurate especially for asymptomatic infection and early infection detection
• Take a nasal fluid sample for testing
Materials and components include: (Note the components, for example: the style of the extraction tube may not be exactly the same as the photo, but it will not affect the functionality)
1. Detection device
3. Disinfecting cotton swabs
4. Antigen extraction tubes for extraction reagents
5. Collection bag
Allow the test device extraction reagents and specimens to equilibrate to room temperature (15~30°C) before testing. Please keep the temperature at 15~30°C and the relative humidity at 20%-80% during the whole test process.
1. Unpack and remove the detection unit. Check the viewing window and sample port (S). Should be used within an hour.
2. Twist the extraction nozzle open.
3. Press the round hole on the top of the box to make a holder for the extraction tube, then insert the bottom of the extraction tube into the hole.
4. Remove the sterile swab from the package.
5. Relax your head. Carefully insert the swab into the nasal cavity, the tip of the swab should be inserted 2 cm until resistance is encountered.
6. Rotate the swab inside the nose for a minimum of 5 turns. Using the same swab, repeat the same process for the other nasal cavity to ensure a sufficient number of samples are collected.
7. Remove the swab from the nasal cavity.
8. Put the swab specimen into the extraction tube, turn the swab for about 10-15 seconds, and press the swab head against the tube wall 3 times to release the antigen in the swab.
9. Tighten the mouth of the extraction tube firmly and twist off the nozzle cap of the extraction tube.
10. Holding the extraction tube vertically, add three drops of test sample to the well (S) and start timing.
11. Please read the result within 15 minutes and within 20 minutes, otherwise the result will be invalid.
12. Please put all used materials in the provided collection bag and dispose of the test kit with your normal household waste.
【Interpretation of test results】
Only (C) has color
If only the control line (C) is colored and the test line (T) is colorless, it means that no COVID-19 (SARS-CoV-2) antigen was detected and the result is negative.
If the test is negative: continue to follow the rules on exposure and protective measures. Even with a negative test result, there may be a risk of infection. If in doubt about the results, please repeat the test after 1-2 days, as the novel coronavirus cannot be accurately detected at all stages of infection.
(C) and (T) are both colored
If both the control line (C) and the test line (T) appear in color, the COVID-19 (SARS-CoV-2) antigen has been detected and the result is positive.
If the test result is positive:
· The user is suspected of being infected with COVID-19 (SARS-CoV-2).
· Contact your doctor or local health department immediately. · Follow local self-isolation guidelines.
· Confirm by PCR test.
(C) is colorless
If the control line (C) is colorless, the test is considered invalid whether or not the test line (T) is visible. A new test needs to be performed with a new test setup.
If the test result is invalid, it may be caused by incorrect test operation. Please repeat the test. If the test results are still invalid, contact your doctor or a COVID-19 (SARS-CoV-2) testing center.
【Storage Conditions and Usage Period】
Store at 2°C~30°C, valid for 24 months. After the aluminum foil bag is opened, the test device should be used as soon as possible within 1 hour (15~30°C, relative humidity ≤80%).
The product 480 pcs, the quantity is too many, not available for Taiwan as the custom will hold your goods.